|Samples||Venous serum, venous whole blood, fingerstick whole blood|
|Measuring range||Linearity between 0.3 - 8 mmol/l detected. Screen can show the following result 10 (higher than 10) mmol/l|
|Certification||CE 0344 marking, for professional use and self-test|
|Accuracy||The internal performance tests have produced the following results:
For venous serum, venous whole blood and venous fingerstick 10% or 0.1 mmol/l for 95% of the measurements.
For venous fingerstick including the fingerstick itself 15% or 0.15 mmol/l for 95% of the measurements with a constant increase of 0.05 mmol/l.
|Shelf life||1 year (indicated on every disposable)|
|Co-medication||No interference found:
Potassium, Ammonium, Calcium, Magnesium, Zinc, Urea, Haemoglobin, Albumin, Creatinine, Lipemic samples, Billirubin, Iron
Sodium: Measurements show an anomaly. The maximum deviation of 0.15 must be taken into account.
Haemolysis: The measurements during the method comparison for the validation studies with fingerstick whole blood include the expected variation of haemolysis. It is assumed that this is the cause of the lower accuracy and the constant increase as stated in the accuracy field.
Other substances: no information known
|Memory capacity||Saves the 100 most recent test results|
|Measuring time||~9 minutes|
|Reagents||Ready for use|
|Acceptable anticoagulant||Potassium EDTA|
|Calibration||No calibration necessary|
|Ambient temperature during operation||15 ˚C - 28 ˚C|
|Humidity during operation||30% - 80%|
|Multireader dimensions||10 x 5 x 20 cm (wxhxl)|
|Transport and storage conditions for Multireader||-20˚C to 60˚C|
|Transport and storage conditions for lab-chip||4˚C to 40˚C|
|Dimensions lab-chip||4 cm x 4 cm x 0,9 cm|
|Weight Multireader||1250 g|
|Weight lab-chip||5,9 g|
|Measuring conditions||for indoor use only|
|Supply Voltage||100 - 240 V|
|Mains frequency||50 - 60 Hz|
|Input current||0,6 A|
|Maximum operating altitude||2000 m|
|Mains voltage fluctuations||+10% tot -10%|
|Pollution Degree Rating||2|